Atomic emission spectroscopy
Atomic emission spectroscopy (AES) is a method of chemical analysis that uses the intensity of light emitted from a flame, plasma, arc, or spark at a particular wavelength to determine the quantity of an element in a sample. The wavelength of the atomic spectral line gives the identity of the element while the intensity of the emitted light is proportional to the number of atoms of the element.
Flame emission spectroscopy
A frequent application of the emission measurement with the flame is the regulation of alkali metals for pharmaceutical analytics.
Inductively coupled plasma atomic emission spectroscopy
Inductively coupled plasma atomic emission spectroscopy (ICP-AES) uses an inductively coupled plasma to produce excited atoms and ions that emit electromagnetic radiation at wavelengths characteristic of a particular element.
Advantages of ICP-AES are excellent limit of detection and linear dynamic range, multi-element capability, low chemical interference and a stable and reproducible signal. Disadvantages are spectral interferences (many emission lines), cost and operating expense and the fact that samples typically must be in solution.
Spark and arc atomic emission spectroscopy
Spark or arc atomic emission spectroscopy is used for the analysis of metallic elements in solid samples. For non-conductive materials, the sample is ground with graphite powder to make it conductive. In traditional arc spectroscopy methods, a sample of the solid was commonly ground up and destroyed during analysis. An electric arc or spark is passed through the sample, heating it to a high temperature to excite the atoms within it. The excited analyte atoms emit light at characteristic wavelengths that can be dispersed with a monochromator and detected. In the past, the spark or arc conditions were typically not well controlled, the analysis for the elements in the sample were qualitative. However, modern spark sources with controlled discharges can be considered quantitative. Both qualitative and quantitative spark analysis are widely used for production quality control in foundries and steel mills.
- Inductively coupled plasma atomic emission spectroscopy
- Atomic absorption spectroscopy
- Atomic spectroscopy
- Laser-induced breakdown spectroscopy
- Stáhlavská A (April 1973). "[The use of spectrum analytical methods in drug analysis. 1. Determination of alkaline metals using emission flame photometry]". Pharmazie (in German) 28 (4): 238–9. PMID 4716605.
- Stefánsson A, Gunnarsson I, Giroud N (2007). "New methods for the direct determination of dissolved inorganic, organic and total carbon in natural waters by Reagent-Free Ion Chromatography and inductively coupled plasma atomic emission spectrometry". Anal. Chim. Acta 582 (1): 69–74. PMID 17386476. doi:10.1016/j.aca.2006.09.001.
- Mermet, J. M. (2005). "Is it still possible, necessary and beneficial to perform research in ICP-atomic emission spectrometry?". J. Anal. At. Spectrom. 20: 11–16. doi:10.1039/b416511j.|url=http://www.rsc.org/publishing/journals/JA/article.asp?doi=b416511j%7Cformat=%7Caccessdate=2007-08-31
- Reynolds, R. J.; Thompson, K. C. (1978). Atomic absorption, fluorescence, and flame emission spectroscopy: a practical approach. New York: Wiley. ISBN 0-470-26478-0.
- Uden, Peter C. (1992). Element-specific chromatographic detection by atomic emission spectroscopy. Columbus, OH: American Chemical Society. ISBN 0-8412-2174-X.
- Atomic Emission Spectroscopy Tutorial
- 16x16px Media related to Atomic emission spectroscopy at Wikimedia Commons