A dye laser is a laser which uses an organic dye as the lasing medium, usually as a liquid solution. Compared to gases and most solid state lasing media, a dye can usually be used for a much wider range of wavelengths, often spanning 50 to 100 nanometers or more. The wide bandwidth makes them particularly suitable for tunable lasers and pulsed lasers. The dye rhodamine 6G, for example, can be tuned from 635 nm (orangish-red) to 560 nm (greenish-yellow), and produce pulses as short as 16 femtoseconds. Moreover, the dye can be replaced by another type in order to generate an even broader range of wavelengths with the same laser, from the near-infrared to the near-ultraviolet, although this usually requires replacing other optical components in the laser as well.
In addition to the usual liquid state, dye lasers are also available as solid state dye lasers (SSDL). SSDL use dye-doped organic matrices as gain medium.
A dye laser consists of an organic dye mixed with a solvent, which may be circulated through a dye cell, or streamed through open air using a dye jet. A high energy source of light is needed to 'pump' the liquid beyond its lasing threshold. A fast discharge flashlamp or an external laser is usually used for this purpose. Mirrors are also needed to oscillate the light produced by the dye’s fluorescence, which is amplified with each pass through the liquid. The output mirror is normally around 80% reflective, while all other mirrors are usually more than 99.9% reflective. The dye solution is usually circulated at high speeds, to help avoid triplet absorption and to decrease degradation of the dye. A prism or diffraction grating is usually mounted in the beam path, to allow tuning of the beam.
Because the liquid medium of a dye laser can fit any shape, there are a multitude of different configurations that can be used. A Fabry–Pérot laser cavity is usually used for flashlamp pumped lasers, which consists of two mirrors, which may be flat or curved, mounted parallel to each other with the laser medium in between. The dye cell is usually side-pumped, with one or more flashlamps running parallel to the dye cell in a reflector cavity. The reflector cavity is often water cooled, to prevent thermal shock in the dye caused by the large amounts of near-infrared radiation which the flashlamp produces. Axial pumped lasers have a hollow, annular-shaped flashlamp that surrounds the dye cell, which has lower inductance for a shorter flash, and improved transfer efficiency. Coaxial pumped lasers have an annular dye cell that surrounds the flash lamp, for even better transfer efficiency, but have a lower gain due to diffraction losses. Flash pumped lasers can be used only for pulsed output applications.
A ring laser design is often chosen for continuous operation, although a Fabry–Pérot design is sometimes used. In a ring laser, the mirrors of the laser are positioned to allow the beam to travel in a circular path. The dye cell, or cuvette, is usually very small. Sometimes a dye jet is used to help avoid reflection losses. The dye is usually pumped with an external laser, such as a nitrogen, excimer, or frequency doubled Nd:YAG laser. The liquid is circulated at very high speeds, to prevent triplet absorption from cutting off the beam. Unlike Fabry–Pérot cavities, a ring laser does not generate standing waves which cause spatial hole burning, a phenomenon where energy becomes trapped in unused portions of the medium between the crests of the wave. This leads to a better gain from the lasing medium.
The dyes used in these lasers contain rather large organic molecules which fluoresce. The incoming light excites the dye molecules into the state of being ready to emit stimulated radiation, the singlet state. In this state, the molecules emit light via fluorescence, and the dye is transparent to the lasing wavelength. Within a microsecond, or less, the molecules will change to their triplet state. In the triplet state, light is emitted via phosphorescence, and the molecules absorb the lasing wavelength, making the dye opaque. Liquid dyes also have an extremely high lasing threshold. Flashlamp pumped lasers need a flash with an extremely short duration, to deliver the large amounts of energy necessary to bring the dye past threshold before triplet absorption overcomes singlet emission. Dye lasers with an external pump laser can direct enough energy of the proper wavelength into the dye with a relatively small amount of input energy, but the dye must be circulated at high speeds to keep the triplet molecules out of the beam path.
Since organic dyes tend to decompose under the influence of light, the dye solution is normally circulated from a large reservoir. The dye solution can be flowing through a cuvette, i.e., a glass container, or be as a dye jet, i.e., as a sheet-like stream in open air from a specially-shaped nozzle. With a dye jet, one avoids reflection losses from the glass surfaces and contamination of the walls of the cuvette. These advantages come at the cost of a more-complicated alignment.
Liquid dyes have very high gain as laser media. The beam needs to make only a few passes through the liquid to reach full design power, and hence, the high transmittance of the output coupler. The high gain also leads to high losses, because reflection from the dye cell walls, or flashlamp reflector, will dramatically reduce the amount of energy available to the beam. Pump cavities are often coated, anodized, or otherwise made of a material that will not reflect at the lasing wavelength while reflecting at the pump wavelength.
CW dye lasers
Continuous-wave (CW) dye lasers often use a dye jet. CW dye-lasers can have a linear or a ring cavity, and provided the foundation for the development of femtosecond lasers.
Narrow linewidth dye lasers
Dye lasers' emission is inherently broad. However, tunable narrow linewidth emission has been central to the success of the dye laser. In order to produce narrow bandwidth tuning these lasers use many types of cavities and resonators which include gratings, prisms, multiple-prism grating arrangements, and etalons.
The first narrow linewidth dye laser, introduced by Hänsch, used a Galilean telescope as beam expander to illuminate the diffraction grating. Next were the grazing-incidence grating designs and the multiple-prism grating configurations. The various resonators and oscillator designs developed for dye lasers have been successfully adapted to other laser types such as the diode laser. The physics of narrow-linewidth multiple-prism grating lasers was explained by Duarte and Piper.
Some of the laser dyes are rhodamine (orange, 540--680 nm), fluorescein (green, 530--560 nm), coumarin (blue 490--620 nm), stilbene (violet 410--480 nm), umbelliferone (blue, 450--470 nm), tetracene, malachite green, and others. While some dyes are actually used in food coloring, most dyes are very toxic, and often carcinogenic. Many dyes, such as rhodamine 6G, (in its chloride form), can be very corrosive to all metals except stainless steel. Although dyes have very broad fluorescence spectrums, the dye's absorption and emission will tend to center on a certain wavelength and taper off to each side, forming a tunability curve, with the absorption center being of a shorter wavelength than the emission center. Rhodamine 6G, for example, has its highest output around 590 nm, and the conversion efficiency lowers as the laser is tuned to either side of this wavelength.
A wide variety of solvents can be used, although some dyes will dissolve better in some solvents than in others. Some of the solvents used are water, glycol, ethanol, methanol, hexane, cyclohexane, cyclodextrin, and many others. Solvents are often highly toxic, and can sometimes be absorbed directly through the skin, or through inhaled vapors. Many solvents are also extremely flammable. The various solvents can also have an effect on the specific color of the dye solution and, thus, on the lasing bandwidth obtainable with a particular laser-pumping source.
Adamantane is added to some dyes to prolong their life.
Cycloheptatriene and cyclooctatetraene (COT) can be added as triplet quenchers for rhodamine G, increasing the laser output power. Output power of 1.4 kilowatt at 585 nm was achieved using Rhodamine 6G with COT in methanol-water solution.
Flashlamps and several types of lasers can be used to optically pump dye lasers. A partial list of excitation lasers include:
- Copper vapor lasers
- Diode lasers
- Excimer lasers
- Nd:YAG lasers (mainly second and third harmonics)
- Nitrogen lasers
- Ruby lasers
- Argon ion lasers in the CW regime
- Krypton ion lasers in the CW regime
Ultra-short optical pulses
R. L. Fork, B. I. Greene, and C. V. Shank demonstrated, in 1981, the generation of ultra-short laser pulse using a ring-dye laser (or dye laser exploiting colliding pulse mode-locking). Such kind of laser is capable of generating laser pulses of ~ 0.1 ps duration.
The introduction of grating techniques and intra-cavity prismatic pulse compressors eventually resulted in the routine emission of femtosecond dye laser pulses.
Dye lasers are very versatile. In addition to their recognized wavelength agility these lasers can offer very large pulsed energies or very high average powers. Flashlamp-pumped dye lasers have been shown to yield hundreds of Joules per pulse and copper-laser-pumped dye lasers are known to yield average powers in the kilowatt regime.
Dye lasers are used in many applications including:
- astronomy (as laser guide stars),
- atomic vapor laser isotope separation
In laser medicine these lasers are applied in several areas, including dermatology where they are used to make skin tone more even. The wide range of wavelengths possible allows very close matching to the absorption lines of certain tissues, such as melanin or hemoglobin, while the narrow bandwidth obtainable helps reduce the possibility of damage to the surrounding tissue. They are used to treat port-wine stains and other blood vessel disorders, scars and kidney stones. They can be matched to a variety of inks for tattoo removal, as well as a number of other applications.
In spectroscopy, dye lasers can be used to study the absorption and emission spectra of various materials. Their tunability, (from the near-infrared to the near-ultraviolet), narrow bandwidth, and high intensity allows a much greater diversity than other light sources. The variety of pulse widths, from ultra-short, femtosecond pulses to continuous-wave operation, makes them suitable for a wide range of applications, from the study of fluorescent lifetimes and semiconductor properties to lunar laser ranging experiments.
Tunable lasers are used in swept-frequency metrology to enable measurement of absolute distances with very high accuracy. A two axis interferometer is set up and by sweeping the frequency, the frequency of the light returning from the fixed arm is slightly different from the frequency returning from the distance measuring arm. This produces a beat frequency which can be detected and used to determine the absolute difference between the lengths of the two arms.
|40x40px||Wikimedia Commons has media related to Dye lasers.|
- Dye Laser Principles: With Applications by Frank J. Duarte, Lloyd W. Hillman -- Academic Press 1990 Page 42
- F. P. Schäfer (Ed.), Dye Lasers (Springer-Verlag, Berlin, 1990).
- F. J. Duarte and L. W. Hillman (Eds.), Dye Laser Principles (Academic, New York, 1990).
- Design and Analysis of Flashlamp Systems for Pumping Organic Dye Lasers – J. F. Holzrichter and A. L. Schawlow. Annals of the New York Academy of Sciences
- Simmer-Enhanced Flashlamp Pumped Dye Laser – T.K. Yee, B. Fan and T.K. Gustafson. Applied Optics – Vol. 18, No. 8
- General Xenon Flash and Strobe Design Guidelines
- Sam's Laser FAQ - Home-Built Dye Laser
- Encyclopedia of Laser Physics and Technology - spatial hole burning, SHB, laser, single-frequency operation
- Laser fundamentals by William T. Silfvast – Cambridge University Press 1996 Page 397-399
- "Principles of Lasers", by Orazio Svelto
- F. P. Schäfer and K. H. Drexhage, Dye Lasers., 2nd rev. ed., vol. 1, Berlin ; New York: Springer-Verlag, 1977
- O. G. Peterson, S. A. Tuccio, B. B. Snavely, CW operation of an organic dye solution laser, Appl. Phys. Lett. 42, 1917-1918 (1970).
- F. J. Duarte and L. W. Hillman, Dye Laser Principles (Academic, New York, 1990) Chapter 4.
- T. W. Hänsch, Repetitively Pulsed Tunable Dye Laser for High Resolution Spectroscopy, Appl. Opt. 11, 895-898 (1972).
- I. Shoshan, N. N. Danon, and U. P. Oppenheim, Narrowband operation of a pulsed dye laser without intracavity beam expansion, J. Appl. Phys. 48, 4495-4497 (1977).
- M. G. Littman and H. J. Metcalf, Spectrally narrow pulsed dye laser without beam expander, Appl. Opt. 17, 2224-2227 (1978).
- F. J. Duarte and J. A. Piper, A double-prism beam expander for pulsed dye lasers, Opt. Commun. 35, 100-104 (1980).
- F. J. Duarte and J. A. Piper, A prism preexpanded grazing incidence pulsed dye laser, Appl. Opt. 20, 2113-2116 (1981).
- P. Zorabedian, Tunable external cavity semiconductor lasers, in Tunable Lasers Handbook, F. J. Duarte (Ed.) (Academic, New York, 1995) Chapter 8.
- F. J. Duarte and J. A. Piper, Dispersion theory of multiple-prism beam expanders for pulsed dye lasers, Opt. Commun. 43, 303–307 (1982).
- Amnon Yariv, Optical Electronics in Modern Communications, Fifth Edition, page 266
- F. J. Duarte and L. W. Hillman (Eds.), Dye Laser Principles (Academic, New York, 1990) Chapters 5 and 6.
- R. L. Fork, B. I. Greene, and C. V. Shank (1981), “Generation of optical pulses shorter than 0.1 psec by colliding pulse mode locking,” Applied Physics Letters, 38: 671–672.
- F. J. Duarte (Ed.), High Power Dye Lasers (Springer-Verlag, Berlin,1991).
- M. A. Akerman, Dye laser isotope separation, in Dye Laser Principles, F. J. Duarte and L. W. Hillman (Eds.)(Academic, New York, 1990) Chapter 9.
- D. Klick, Industrial applications of dye lasers, in Dye Laser Principles, F. J. Duarte and L. W. Hillman (Eds.)(Academic, New York, 1990) Chapter 8.
- W. Demtröder, Laser Spectroscopy, 3rd Ed. (Springer, 2003).
- L. Goldman, Dye lasers in medicine, in Dye Laser Principles , F. J. Duarte and L. W. Hillman, Eds. (Academic, New York, 1990) Chapter 10.
- A. Costela et al., Medical applications of dye lasers, in Tunable Laser Applications, F. J. Duarte (Ed.), 2nd Ed. (CRC, New York, 2009) Chapter 8
- Tunable Laser Applications By Frank J. Duarte – CRC Press 2009 Page 227-235
- The Laser Guidebook By Jeff Hecht – McGraw Hill 1992 Page 294
- Highly linear, Widerange Swept Frequency Generation at Microwave and Optical Frequencies