## Frequent Links

# Multiphoton intrapulse interference phase scan

This article provides insufficient context for those unfamiliar with the subject. (October 2009) |

**Multiphoton intrapulse interference phase scan** (**MIIPS**) is a method used in ultrashort laser technology that simultaneously measures (phase characterization), and compensates (phase correction) femtosecond laser pulses using an adaptive pulse shaper.
Current ultrashort laser pulse characterization methods (streak camera, autocorrelation, FROG, SPIDER etc.) can only measure the pulse characteristics. Thus, the application of an ultrashort pulse is limited as the electromagnetic field of a pulse is determined by the laser cavity, and varies dramatically when the pulse duration is in the femtosecond region. It is therefore highly desirable to have a method which can not only characterize the pulse, but also correct the pulse to specific shapes for various applications in which repeatable pulse characteristics are requested. MIIPS can not only measure the pulse but also correct the high-order dispersion, thus is highly preferable for applications where repeatable electromagnetic field is important, such as to generate ultrashort pulses which are transform limited or possess specific phase characteristics.

## Contents

## Theory

A MIIPS-based device consists of two basic components controlled by a computer: a pulse shaper (usually a liquid crystal based spatial light modulator - SLM) and a spectrometer. The pulse shaper allows manipulation of the spectral phase and/or amplitude of the ultrashort pulses. The spectrometer records the spectrum of a nonlinear optical process such as second harmonic generation produced by the laser pulse. The MIIPS process is analogous to the Wheatstone bridge in electronics. A well-known (calibrated) spectral phase function is used in order to measure the unknown spectral phase distortions of the ultrashort laser pulses. Typically, the known superimposed function is a periodic sinusoidal function that is scanned across the bandwidth of the pulse.

MIIPS is similar to FROG in that a frequency trace is collected for the characterization of the ultrashort pulse. In Frequency-resolved optical gating, a FROG trace is collected through scanning the ultrashort pulse across the temporal axis, and detecting the spectrum of the nonlinear process. It can be expressed as

- <math>

I(\omega,\tau)=\left|\int{E(t)g(t-\tau)e^{i\omega t}\mathrm{d}t}\right|^2 </math>

In MIIPS, instead of scanning on the temporal domain, a series of phase scan is applied on the phase domain of the pulse. The trace of the MIIPS scan consists of the second-harmonic spectra of each phase scan. The signal of MIIPS can be written as

- <math>

I(2\omega)=\left|\int{|E(\omega)|^2e^{i\phi}\mathrm{d}\phi}\right|^2 </math>

The phase scan in MIIPS is realized with introducing a well-known reference function, <math>f(\omega)</math>, by the pulse shaper to locally cancel distortions by the unknown spectral phase, <math>\Phi(\omega)</math>, of the pulse. The sum of the unknown phase and the reference phase is given by <math>\phi(\omega)=\Phi(\omega)+ f(\omega)</math>. Because the frequency doubled spectrum of the pulse depends on <math>\phi(\omega)</math>, it is possible to accurately retrieve the unknown <math>\Phi(\omega)</math>.

The phase modulation procedure of the physical process is generally a continuous function. Thus, the SHG signal can be expanded with a Taylor expansion around <math>\omega</math>:

- <math>

I(\omega)= \left| \int| E(\omega+\Omega)| |E(\omega-\Omega)| \times \text{exp} \{i[\phi(\omega+\Omega)+\phi(\omega -\Omega)]\} \mathrm{d}\Omega \right|^2 </math>

And

- <math>

\phi(\omega+\Omega)+\phi(\omega-\Omega)=2\phi0+\phi*(\omega)\Omega^2+...+\frac{2}{(2n)!}\phi^{2n'}(\omega)\Omega^{2n}*
</math>

According to this equation, the SHG signal reaches maximum when <math>\phi(\omega+\Omega)+\phi(\omega-\Omega)</math> is zero. This is equivalent to <math>\Phi*(\omega)=-f*(\omega)</math>. Through scanning of <math>f(\omega)</math>, the <math>\Phi(\omega)</math> can be decided.

The frequency doubled spectrum recorded for each full scan of the reference phase <math>4(\pi)</math> results in two replicas of the MIIPS trace (see Figure 1, four replicas shown). From this data, a 2D plot for SHG(<math>\omega,\omega</math>) is constructed where <math>\omega=\pi c/\lambda_{SHG}</math>. The second harmonic spectrum of the resulting pulse has a maximum amplitude at the frequency where the second derivative of the pulse has been compensated. The lines describing <math>\omega_{m}(\omega)</math> are used to obtain analytically the second derivative of the unknown phase. After double integration the phase distortions are known. The system then introduces a correction phase to cancel the distortions and achieve shorter pulses. The absolute accuracy of MIIPS improves as the phase distortions diminish, therefore an iterative procedure of measurement and compensation is applied to reduce phase distortions below 0.1 radian for all frequencies within the bandwidth of the laser.

When all phase distortions have been eliminated, the pulses are the shortest they can be and are considered to be Bandwidth-limited-pulse|transform limited (TL). The MIIPS trace corresponding to TL pulses shows straight parallel lines separated by <math>\pi</math>. Once spectral phase distortions have been eliminated, the shaper can be used to introduce calibrated phases and amplitudes to control laser induced processes.

MIIPS technology has been applied successfully in selective excitation of multiphoton imaging and femtosecond light-mass interaction study.

## Experimental setup

The expanded laser beam reaches the Diffractive grating (G) first, the first-order reflection is deflected to the Mirror (M) and then to the Curved Mirror (CM). The Curved Mirror reflects the laser to the Spatial Light Modulator (SLM). The phases are applied through the Spatial Light Modulator to each component of the frequency. The laser is then retro-reflected. By using a nonlinear medium, the nonlinear (SHG, THG, etc.) spectra vs. the phase scan can be recorded as a MIIPS trace for the characterization of the pulse. Once the pulse is characterized, a compensatory phase can be applied to the ultrashort pulse through the SLM.

## Other ultrashort pulse measurement techniques

- Frequency-resolved Optical Gating (FROG)
- Spectral phase interferometry for direct electric-field reconstruction (SPIDER)

## References

- M. Dantus, V. V. Lozovoy, and I. Pastirk, "Measurement and Repair: The femtosecond Wheatstone Bridge." OE Magazine 9 (2003).
- V. V. Lozovoy, I. Pastirk, and M. Dantus, "Multiphoton intrapulse interference 4: Characterization and compensation of the spectral phase of ultrashort laser pulses." Optics Letters 29, 775-777 (2004).
- B. Xu, J. M. Gunn, J. M. Dela Cruz, V. V. Lozovoy, M. Dantus, “Quantitative investigation of the MIIPS method for phase measurement and compensation of femtosecond laser pulses,” J. Optical Society of America B 23, 750-759 (2006).