Open Access Articles- Top Results for Spectrophotometry
Research & Reviews: Journal of Pharmaceutical AnalysisDual Wavelength Spectrophotometric Method for Estimation of Cilnidipine and Telmisartan in Their Combined Dosage Form.
Journal of Forensic ResearchUltraviolet-Visible Spectrophotometry (UV-VIS) and SALIgAE® Qualitative and Semi-quantitative Tools for the Analysis of Salivary Amylase
Journal of Oral Hygiene & HealthA Color Analysis of Smokers Melanosis Using a Non-Contact Type Dental Spectrophotometer
Journal of Analytical & Bioanalytical TechniquesOrganised Surfactants Assemblies and Spectrophotometry: Application to The Analysis of Palladium (II) in Different Matrices
Journal of Bioequivalence & BioavailabilityBioequivalence Study of Two 10 mg Montelukast Immediate-Release Tablets Formulations: A Randomized, Single-Dose, Open-Label, Two Periods, Crossover
|16px Jacqueline Boytim discusses the development and early use of spectrophotometers, Chemical Heritage Foundation|
In chemistry, spectrophotometry is the quantitative measurement of the reflection or transmission properties of a material as a function of wavelength. It is more specific than the general term electromagnetic spectroscopy in that spectrophotometry deals with visible light, near-ultraviolet, and near-infrared, but does not cover time-resolved spectroscopic techniques.
Spectrophotometry uses photometers that can measure a light beam's intensity as a function of its color (wavelength) known as spectrophotometers. Important features of spectrophotometers are spectral bandwidth, (the range of colors it can transmit through the test sample), and the percentage of sample-transmission, and the logarithmic range of sample-absorption and sometimes a percentage of reflectance measurement.
A spectrophotometer is commonly used for the measurement of transmittance or reflectance of solutions, transparent or opaque solids, such as polished glass, or gases. However they can also be designed to measure the diffusivity on any of the listed light ranges that usually cover around 200 nm - 2500 nm using different controls and calibrations. Within these ranges of light, calibrations are needed on the machine using standards that vary in type depending on the wavelength of the photometric determination.
An example of an experiment in which spectrophotometry is used is the determination of the equilibrium constant of a solution. A certain chemical reaction within a solution may occur in a forward and reverse direction where reactants form products and products break down into reactants. At some point, this chemical reaction will reach a point of balance called an equilibrium point. In order to determine the respective concentrations of reactants and products at this point, the light transmittance of the solution can be tested using spectrophotometry. The amount of light that passes through the solution is indicative of the concentration of certain chemicals that do not allow light to pass through.
The use of spectrophotometers spans various scientific fields, such as physics, materials science, chemistry, biochemistry, and molecular biology. They are widely used in many industries including semiconductors, laser and optical manufacturing, printing and forensic examination, as well in laboratories for the study of chemical substances. Ultimately, a spectrophotometer is able to determine, depending on the control or calibration, what substances are present in a target and exactly how much through calculations of observed wavelengths.
There are two major classes of devices: single beam and double beam. A double beam spectrophotometer compares the light intensity between two light paths, one path containing a reference sample and the other the test sample. A single-beam spectrophotometer measures the relative light intensity of the beam before and after a test sample is inserted. Although comparison measurements from double-beam instruments are easier and more stable, single-beam instruments can have a larger dynamic range and are optically simpler and more compact. Additionally, some specialized instruments, such as spectrophotometers built onto microscopes or telescopes, are single-beam instruments due to practicality.
Historically, spectrophotometers use a monochromator containing a diffraction grating to produce the analytical spectrum. The grating can either be movable or fixed. If a single detector, such as a photomultiplier tube or photodiode is used, the grating can be scanned stepwise so that the detector can measure the light intensity at each wavelength (which will correspond to each "step"). Arrays of detectors, such as charge coupled devices (CCD) or photodiode arrays (PDA) can also be used. In such systems, the grating is fixed and the intensity of each wavelength of light is measured by a different detector in the array. Additionally, most modern mid-infrared spectrophotometers use a Fourier transform technique to acquire the spectral information. The technique is called Fourier transform infrared spectroscopy.
When making transmission measurements, the spectrophotometer quantitatively compares the fraction of light that passes through a reference solution and a test solution. For reflectance measurements, the spectrophotometer quantitatively compares the fraction of light that reflects from the reference and test samples. Light from the source lamp is passed through a monochromator, which diffracts the light into a "rainbow" of wavelengths and outputs narrow bandwidths of this diffracted spectrum. Discrete frequencies are transmitted through the test sample. Then the photon flux density (watts per metre squared usually) of the transmitted or reflected light is measured with a photodiode, charge coupled device or other light sensor. The transmittance or reflectance value for each wavelength of the test sample is then compared with the transmission or reflectance values from the reference sample.
In short, the sequence of events in a modern spectrophotometer is as follows:
- The light source is imaged upon the sample.
- A fraction of the light is transmitted or reflected from the sample.
- The light from the sample is imaged upon the entrance slit of the monochromator.
- The monochromator separates the wavelengths of light and focuses each of them onto the photodetector sequentially.
Many older spectrophotometers must be calibrated by a procedure known as "zeroing." The absorbency of a reference substance is set as a baseline value, so the absorbencies of all other substances are recorded relative to the initial "zeroed" substance. The spectrophotometer then displays % absorbency (the amount of light absorbed relative to the initial substance).
Visible region 400–700 nm spectrophotometry is used extensively in colorimetry science. It is a known fact that it operates best at the range of 0.2-0.8 O.D. Ink manufacturers, printing companies, textiles vendors, and many more, need the data provided through colorimetry. They take readings in the region of every 5–20 nanometers along the visible region, and produce a spectral reflectance curve or a data stream for alternative presentations. These curves can be used to test a new batch of colorant to check if it makes a match to specifications, e.g., ISO printing standards.
Traditional visible region spectrophotometers cannot detect if a colorant or the base material has fluorescence. This can make it difficult to manage color issues if for example one or more of the printing inks is fluorescent. Where a colorant contains fluorescence, a bi-spectral fluorescent spectrophotometer is used. There are two major setups for visual spectrum spectrophotometers, d/8 (spherical) and 0/45. The names are due to the geometry of the light source, observer and interior of the measurement chamber. Scientists use this instrument to measure the amount of compounds in a sample. If the compound is more concentrated more light will be absorbed by the sample; within small ranges, the Beer-Lambert law holds and the absorbance between samples vary with concentration linearly. In the case of printing measurements two alternative settings are commonly used- without/with uv filter to control better the effect of uv brighteners within the paper stock.
Samples are usually prepared in cuvettes; depending on the region of interest, they may be constructed of glass, plastic (visible spectrum region of interest), or quartz (Far UV spectrum region of interest).
- Estimating dissolved organic carbon concentration
- Specific Ultraviolet Absorption for metric of aromaticity
- Bial's Test for concentration of pentoses
Spectrophotometers designed for the infrared region are quite different because of the technical requirements of measurement in that region. One major factor is the type of photosensors that are available for different spectral regions, but infrared measurement is also challenging because virtually everything emits IR light as thermal radiation, especially at wavelengths beyond about 5 μm.
Another complication is that quite a few materials such as glass and plastic absorb infrared light, making it incompatible as an optical medium. Ideal optical materials are salts, which do not absorb strongly. Samples for IR spectrophotometry may be smeared between two discs of potassium bromide or ground with potassium bromide and pressed into a pellet. Where aqueous solutions are to be measured, insoluble silver chloride is used to construct the cell.
Spectroradiometers, which operate almost like the visible region spectrophotometers, are designed to measure the spectral density of illuminants. Applications may include evaluation and categorization of lighting for sales by the manufacturer, or for the customers to confirm the lamp they decided to purchase is within their specifications. Components:
- The light source shines onto or through the sample.
- The sample transmits or reflects light.
- The detector detects how much light was reflected from or transmitted through the sample.
- The detector then converts how much light the sample transmitted or reflected into a number.
- Atomic Absorption Spectrophotometry
- Atomic emission spectroscopy
- Inductively coupled plasma atomic emission spectroscopy
- Inductively coupled plasma mass spectrometry
- Slope spectroscopy
- Allen, D., Cooksey, C., & Tsai, B. (2010, October 5). Spectrophotometry. Retrieved from http://www.nist.gov/pml/div685/grp03/spectrophotometry.cfm
- Schwedt, Georg. (1997). The Essential Guide to Analytical Chemistry. (Brooks Haderlie, trans.). Chichester, NY: Wiley. (Original Work Published 1943). pp. 16-17
- Rendina, George. Experimental Methods in Modern Biochemistry W. B. Saunders Company: Philadelphia, PA. 1976. pp. 46-55
|40x40px||Wikimedia Commons has media related to Spectrophotometry.|